Wisman High Voltage Power Supply Co., Ltd was more than once successfully repaired many X-ray generators of XRF Venus200/Venus100 which is producted by Holland Panalytical(formerly Holland Philips analytical). X-ray generator Ouput Power 200W max, Current 4mA max,Voltage 50KV max.
Wisman remind most Panalytical or other instrument users: Aftermore than 20 hoursdown re-boot,please do the aging work for X-ray tube.Closing down 20-100 hours, you should click fast breed key in manual control system,Meanwhile,start HV key and ageing fast. If down 100 hours, Click normal breed key in manual control system, activate HV key and ageing normal.Try to avoid power system sudden failure,Do routine maintenance and keep the instrument placed in laboratories and studios temperature and humidity,environment temperature initiative below 30 degrees, humidity below 80%, so as to avoid unnecessary fault, affecting production and experiment.
Statement: This post is only the personal view of the author and does not represent the opinions of ALLPCB.com.